ESR Fast Detector
ESR Tube: 3.8% of Sodium Citrate (1:4)
1. ESR Tube,1.6ml, 8x120mm
0.4ml 3.8% of Sodium Citrate+1.6ml Blood
2. ESR Tube,1.28ml, 8x120mm
0.32ml 3.8% of Sodium Citrate+1.28ml Blood
Tube Store and Testing Condition:
Store and Testing temperature: 4 to 25 ℃
Within 3 hours to finish the testing.
NOTE: Avoid exposure to direct sunlight. Exceeding the maximum recommended storage temperature may lead to impairment of the tube quality (i.e. vacuum loss, colouring, etc.)
1. Put the ESR detector on the level place.
2. Using the ESR tube to collect the blood and mixing well. (Turn over 6-7times)
3. Put the tube upright on the ESR detector. The tube blue graduation line flush with rack top.
4. Wait for 30mins, align the plasma concave in ESR tube to the zero scale of the ESR detector, read the data by aligning the upper surface of erythrocyte to the scale on the ESR detector.
Better compatibility compared with westergren method
A safe, reliable and airtight operation during the whole process from blood collection to test, free of biological contamination
Designed with spirit level of patent technology, which ensures the detector horizontal and aclinic.
For 10 channels, synchronous operation allowd.
Only 30minutes needed for reading, easy and swift
Equipped with level calibration device, convenient for level correction
|8160ESR||8*120mm||1.6ml||3.8% Sodium Citrate||100||1200|
|810128R||ESR Fast Detector||For 1.28ml||1||10|
|810160R||ESR Fast Detector||For 1.6ml||1||10|